The AMW Series is a great durable and compact pocket scale for those who are seeking the on the go high precision portable scale. The backlit LCD display helps make the numbers viewable and easy to read. The intuitive protective cover provides protection for the scale. With the smooth stainless steel weighing surface clean up is easy. Buttons on the scale give you full control such as: switching modes grams, ounces, carats, grains and tarring. This checkbook size scale offers great range from 55 grams to as little as 0.01 of a gram. You get all this in one scale backed up with our incredible 10 year warranty!
Whether this fits your style or needs - the AMW Series is the portable pocket scale that packs punch and gets the job done!
- 55g Capacity
- 0.01g Resolution
- Backlit LCD
- Flip-open lid protects the delicate weighing surface.
- Great for precision weighing on the go.
- Reads in g, oz, ct, and gn
- Stainless steel weighing platform.
|Capacity||55g / 1.94oz / 275ct / 848.8gn |
|Readability||0.01g / 0.005oz / 0.05ct / 0.2gn |
|Display||5 digit backlit LCD |
|Calibration weight ||50g NOT included |
|Platform dimensions ||2.5 x 3 |
|Scale dimensions ||5 x 3 x 0.75 |
|Power||2 X AAA batteries included|
|Gross weight ||0.47lb|
Thank you for purchasing the American Weigh AMW-Series digital scale. Please read all operating instructions carefully before use. This electronic scale is a precision instrument. With normal care and proper treatment, it will provide years of reliable service.
Never load the scale with more than the maximum capacity. Overloading will permanently damage it!
Avoid any exposure to extreme heat or cold. Your scale works better when operated at normal room temperature. Always allow the scale to acclimate to normal room temperature for at least one hour before use.
Give your scale sufficient warm up time. Usually 30-60 seconds before calibration to give the internal components a chance to stabilize.
Keep your scale in a clean environment. Dust, dirt, moisture, vibration, air currents and a close distance to other electronic equipment can all cause an adverse effect on the reliability and accuracy of your scale.
Handle with care. Gently apply all items to be weighed onto tray top. Although this scale is designed to be quite durable, try to avoid rough treatment as this may permanently damage the internal sensor.
Avoid shaking, dropping or otherwise shocking the scale. This is a precision instrument and must be handled with extreme care.
Only operate the scale on a stable, vibration free surface.
Place Scale on a horizontal flat surface.
1. Press [ON/OFF] key.
2. Wait until "0.0" is displayed.
3. Put the object(s) on the weighing platform.
4. Using the [MODE] key, you can switch between the weighing units.
II. TARE WEIGHING
1. Turn on scale as described above.
2. Place the "tare item" on the platform.
3. Press [TARE] and wait until "0.0" is displayed.
4. Add the "net-weight-item".
Calibration is NOT required before first use. Only calibrate if scale is not weighing properly.
**Make sure you use the correct calibration weight for your scale or the calibration will not be accurate**AMW-55 needs a 50g calibration weight 1. Turn OFF the scale.
2. Press [ON/OFF] shortly. Then immediately press and hold [MODE] until "CAL" is displayed. Release the [MODE] key.
3. Press [MODE] again. Now the display will show "CAL" and "50.00" will flash 3 times.
4. Place a 50g weight on the center of platform. Wait 3-4 seconds. The LCD display will read "PASS", then "50.00".
5. Remove the weight and press [ON/OFF] to turn off the scale. Finished! Repeat if needed.
Please check the availability of weights at the store you bought your scale from.
The primary reasons for inaccuracy or malfunction are low batteries, incorrect calibration, overload or operating on an unstable surface. Please keep this in mind and maintain and operate your scale properly. This scale is a precise instrument and must be handled with the utmost care and caution.
[Display] Elimination of errors
Nothing, Battery-Symbol, [lo] or  Change the batteries.
[OUE2] Recalibrate the scale as shown above.
[EEEEE] You are overloading the scale. Remove the excessive load immediately!
[UNST] Use the scale in a more stable position
[LLLLL] The delicate weighing load cells have been damaged by misusing the scale (i.e. it was shocked, dropped or overloaded). You can try recalibrating the scale. If the load cell has not been damaged too much it will work again after recalibration.
This product was added to our catalog on Wednesday 11 October, 2006.